Views: 2 Author: Site Editor Publish Time: 2021-08-26 Origin: Site
1. There is scattered current interference in the measured pipeline.
2. The tested pipeline is plastic or coated with insulating layer.
Accurate and stable measurement is based on the complete grounding of the magnetic flowmeter. The flowmeter with grounding ring eliminates the interference caused by dispersed current in the pipeline and can provide more reliable flow data.
1. If the electromagnetic flowmeter is installed on plastic pipe or painted pipe, grounding rings must be installed at both ends of the sensor. Otherwise, the electromagnetic flowmeter cannot work normally.
2. There are many materials for grounding ring, such as stainless steel, Hastelloy, titanium, tantalum and platinum.
ARTang is a global leader in measurement instrumentation, services and solutions for industrial process engineering that energizes the transformation of society and industry to achieve a more productive, sustainable future. We provide process solutions for flow, level, pressure, analytics and digital communications, optimizing processes in terms of economic efficiency, safety & environmental impact.
Sign up for our newsletter to receive the latest news.